Boldizar M., *Pelikan K., **Repel D., Vidrickova P., Sevcik A.
University of Veterinary Medicine, Clinic of Surgery, Orthopaedic & Radiology, Komenskeho 73, 041 81 Kosice, Slovak Republic
*St. Anna University Hospital, Clinic of Anaesthesiology & IC, Brno, Czech Republic
**Capio Horton NHS Treatment Centre, Banbury, Oxfordshire, UK
Correspondence address: email@example.com,
Background and Goal of Study:
Airway management during general anaesthesia [GA] is a cardinal problem in the veterinary anaesthesia as well. Open airways can be achieved by a change of a patient’s head position, a face mask [FM] or by an insertion of an endotracheal tube [ETT]. These techniques have advantages and disadvantages. The introduction of the laryngeal mask [LMA] into human anaesthesia was a revolutionary breakthrough at the end of the 20th century which solved problems connected with the FM and ETT. The first use of LMA Classic™ in the veterinary anaesthesia dates back to 1991 and LMA ProSeal™[PLMA] to 2005.
Materials and Methods:
The use of PLMA on animals n=49 in three species [dogs 35, sheep 7, swines 7] was studied. In all the animals Propofol without myorelaxants for the induction into GA, the cuff pressure 40cm H2O, lubricant jelly without a local anaesthetic were used. Ventilation time [VT] through the PLMA, the signs of the gastric reflux [GR], successful PLMA insertion on the first attempt [SIFA] and the malposition [MP] were followed up. The proper position of PLMA, the position of its cuff in the hypopharynx was check by X Ray. The proper position of the distal aperture of the drain tube [DT] was confirmed by insertion of a gastric tube into oesophagus via DT. Three insertion techniques recommended for human anaesthesia and a bubble test were tested.
The three insertion techniques recommended in human anaesthesia are unsuitable to use in a veterinary anaesthesia. The bubble test should be modified because performing a suprasternal checking in animals is impossible. Cuff pressure mean values, with recommended maximum inflation volume, were: dogs 46 ±1,58 SD cmH2O; sheep 46,17 ±0,9 SD cmH2O; swine 46,5 ±1,12 SD cmH2O. Statistical data are shown in the table:
|n||VT Ø min||GR %||SIFA %||MP %|
|DOG||35||76,43 ± 45,56 SD||8,6||91,4||5,7|
|SHEEP||7||50,71 ± 30,99 SD||71,4||100||0|
|SWINE||7||115,71 ± 41,98 SD||0,0||100||0|
The use of PLMA in the veterinary anaesthetic practice significantly improves airway management in animals during GA. On the basis of our results, it is possible to apply model PLMA designed for humans in the airway management in veterinary anaesthesia in these species: dogs, sheep, swine and roe deer. Airway protection in species having a high risk of
regurgitation, like sheep, is excellent and continual regurgitation is easy to control via the PLMA drain tube.
Euroanaesthesia 2007, Munich, Germany, 9-12 June 2007
Necessary cookies are absolutely essential for the website to function properly. This category only includes cookies that ensures basic functionalities and security features of the website. These cookies do not store any personal information.
Any cookies that may not be particularly necessary for the website to function and is used specifically to collect user personal data via analytics, ads, other embedded contents are termed as non-necessary cookies. It is mandatory to procure user consent prior to running these cookies on your website.